|Place of Origin:||China|
|Minimum Order Quantity:||1|
|Detector:||Uncooled Vox FPA||Pixel:||384×288|
|Pixel Size:||17um||Spectral Band:||8~14um|
|Resolution:||1920(H)×1080(V)||Focal Length:||4.7mm-94mm Continuous Zoom|
|Size:||Progressive Scanning 1/2.8"CMOS||Displayed Temperature Range:||(30~50.0)ºC|
JH-2005B Online IR Human Temperature Checking System
1.IR detector: Uncooled Vox FPA
2. Pixel: 384×288
3. Pixel size: 17um
4.Spectral band: 8~14um
5. Bad pixel: <0.05%
7. Day light camera
9. Focal length:4.7mm-94mm continuous zoom
10. Size:progressive scanning 1/2.8"CMOS
11.Displayed temperature range:(30~50.0)ºC
12.Temperature resolution ≤0.1ºC
13.Alarming temperature measurement error:±0.3ºC(Under the specified measuring distance and application environment, when measuring the black body radiation source with diameter of 60mm )
15.Measurement mode: calculate body temperature by measuring body surface temperature within the specified range of ambient temperature; When the warning temperature is not reached, the highest measured value is displayed and the measured point is indicated. When the warning temperature is reached, multiple higher measured values can be indicated and the warning target position can be indicated
16.Calibration mode: a mode for measuring and displaying the temperature of a blackbody source (obtained by conversion)
17.Warning value correction function: it has modifiable indication value correction function
18.It has built-in and external black body automatic correction function.
19. Image output:Video and images can be sent to the remote control room.
20.Working temperature:(10~40)ºC ,humidity:(5~95)%RH .
21. Climatic and environmental adaptability warning temperature measurement error complies with the climatic and environmental requirement in 4.3.3 of GB/T 19870-2005.It requires working temperature within(10~40)ºC, storage and transportation within(-20~60)ºC ,RH(5~95)% (at 30ºC), storage and transportation ≤90 % (at 40ºC)
22.Approval certificate issued by the national quality inspection department shall be provided.
23.The manufacturer promises to upgrade or upgrade the Camera System for free
24.Installation standard: complete wiring on user requirements; install the host of the infrared camera in the position specified by the user, complete hanging installation to work normally and measure temperature.